CTL200 User GuideLast updated: January 17, 2019
The CTL200-1 / CTL200-2 is compatible with Type 1 / Type 2 butterfly laser diodes. Note that it is only compatible with floating diodes, which means that neither the anode nor the cathode should be connected to the case.
The CTL200 is also available in a screw terminal version. Please refer to the CTL200-0 user guide for information specific to this version.
The board must be supplied with the 5.9 V / 3 A power supply that comes with the CTL200. Another power supply can be used, as long as the voltage is between 5.7 V and 6.5 V.
The serial interface can be accessed either via the micro USB connector or via the TX and RX pins of the programming connector.
The serial interface is the only way of configuring the CTL200 operating parameters. Once the configuration is done, it can be saved to the internal memory with the save command. If the configuration switch SW1 is ON at start up, the CTL200 will load the user-defined configuration and the serial interface is no longer needed.
The CTL200 can be controlled directly from a serial port terminal (e.g. Teraterm on Windows) with the following configuration:
- Baud rate: 115200
- Parity: None
- Bits: 8
- Stopbits: 1
- Flow control: None
Example use of the serial port terminal
>> >>version V0.1 >>rtset 10000.000000 >>rtact 10000.023438 >>rtset 12000 12000.000000 >>rtact 11999.853516
Graphical User Interface
A Graphical User Interface is available for Windows. The installer can be downloaded at this link. The GUI provides an effective mean of tuning the PID gains of the temperature controller.
lason: Should be set to 1 to enable the laser current (default value: 0)
ilaser: Laser current in mA (default value: 0.0).
rtset: Thermistor resistance setpoint in Ω (default value: 10000.0).
pgain: Proportional gain of the temperature controller (default value: 0.001).
igain: Integral gain of the temperature controller (default value: 0.0001).
dgain: Differential gain of the temperature controller (default value: 0.005).
rtmin: Minimum thermistor resistance in Ω. Below this value, the laser is automatically disabled (default value: 5000.0).
rtmax: Maximum thermistor resistance in Ω. Above this value, the laser is automatically disabled (default value: 15000.0).
vtmin: Minimum TEC voltage in V (default value: -2.0).
vtmax: Maximum TEC voltage in V (default value: 2.0).
version: Return the firmware version (e.g. V0.1)
lason: Laser status (0 when disabled, 1 when enabled)
vlaser: Laser voltage in V.
iphd: Photodiode current in mA (maximum value: 2.5 mA).
rtact: Actual value of the thermistor resistance in Ω.
itec: TEC current in A.
vtec: TEC voltage in V.
ain1: Voltage on pin AIN1 in V.
ain2: Voltage on pin AIN2 in V.
- Switch 1: When SW1 is ON, the user configuration is loaded at start-up. When SW1 is OFF, the default configuration is loaded at start up.
- Switch 2: SW2 is used for firmware update. It should be kept OFF for normal operation.
- STATUS: Always ON during normal operation
- LASER: Turns ON when the photodiode current is above 10 µA.
- TEMP: Turns ON when the thermistor resistance is within 1 Ω of the setpoint.
The CTL200 has two current modulation inputs available on SMA connectors:
DC modulation input for modulation between DC and 10 MHz. Modulation range is ±1 V. A jumper allows to choose between 3 modulation gains (250 µA/V, 2.5 mA/V or 25 mA/V for the 200 mA version).
AC modulation input for modulation above 100 kHz. Modulation range is ±5 V. Modulation gain is 20 mA/V.
The TEC voltage limits (vtmin and vtmax) and the thermistor resistance limits (rtmin and rtmax) provide an effective way of protecting the laser against thermal runaway. For reliable operation, the aluminium cooling base plate must be properly heatsunk.